One Touch, Friend Forever
contact sales at SINOWON.
Firstly, it must be known what material is tested, please check the typical application in the following table, it will be helpful to choose right probe. If there is coating or plating on the specimen, it is better to choose the motorzied probe with minimum testing force, but you must consider the roughenss of suface. If the roughness is high, the light force probe is not suitable.
Secondly, if the speimen is small and massive, it is better to choose a stationary stand to make the testing reliably.
Thridly, if the specimen is a cyliner-shape, it is better to fix the probe in a small/ large cylinder kit, or MU-200 portable stand, it will be benifical to get a credible testing result.
Fourth, if the specimen can not be reached by the standard length probe, it is able to test it in the lengthen manual porbe or motorized probes. If it is inside wall of some turbines or pipes, it is able to use the manual shorten probe to complete the testing.
|Force||Model||Control||Probe Style||Stands and Kits||Typical Applications|
Standard probe cap
Deep-hole probe cap
Small forgings, cast material,
weld inspection, HAZ
Induction hardened or
carburized machine parts,
turbine weld inspection, HAZ
Measurement in grooves, on gear tooth flanks and roots, turbine blades
Ion-nitrided stamping dies and molds, forms, presses,thin walled parts
Bearings, tooth flanks, turbine blades, inside wall of pipes with Ø >110 mm
Finished precision parts,
gears, bearing raceway
Thin layers, for example, copper or chromium on steelcylinders;
Copper rotogravure cylinders;
Coatings, case hardened parts
Hello, please leave your name and email here before chat online so that we won't miss your message and contact you smoothly.